The QuickSee is a portable, low-cost autorefractor based on wavefront aberrometry—a vision care technology that scan for abnormalities in the eyeball’s optical power. As of August 2017, this product is in the manufacturing stage and is expected to launch Fall 2017. Interview with representative
Testing has been conducted in India, but the goal is to reach many low-resource countries around the world, as well as the United States and other high-resource settings. Interview with representative
PlenOptika has partnered with a reputable low-cost ophthalmic manufacturer in India called Aurolab. In the US, PlenOptika will sell and distribute the Quicksee. Interview with representative
Plenoptika will sell two models of the QuickSee. One will be marketed for high-resource settings, while the other one will be marketed for middle and low-resource settings. The latter model will sell for around 2,000 USD. Interview with representative
Goal 3:: to promote and improve health and well-being.
Optometrists, doctors, and eye care professionals. At times, non-health care professionals
Mass produced. PlenOptika has partnered with a reputable low-cost ophthalmic manufacturer called Aurolab
Doctors, hospitals, and NGOs that focus on eye care can procure a Quicksee directly from Aurolab, PlenOptika’s manufacturing partner, or from a sales link on the PlenOptika website. Interview with representative
Not applicable. As of August 2017, this product is in the manufacturing stage and is expected to launch Fall 2017. Interview with representative
QuickSee works by shining a light into the eye, which generates a point of light on the retina at the back of the eye. The light then bounces off the retina, and the shape of the light as it exits the eye relays information about aberrations, or distortions, that are present.
To identify a patient’s prescription, a special camera called a “wavefront sensor” measures the shape of the light that comes out of the eye. Unlike a conventional camera which takes a single image, a wavefront sensor photographs the eye in segments and analyzes aberrations within each region of the eye. A computer in the device then processes all of the images collectively and calculates any refractive errors that might be present. With that, it is able to approximate a prescription in about 10 seconds per eye, which the QuickSee displays on-screen. The device can also transmit the results via Bluetooth to a smartphone or printer
The Quicksee performs refractive errors tests. It measures sphere cylinder and axis for each eye and takes approximately 10 seconds per eye. The Quicksee uses waveform abberometry. It is a battery powered device and lasts 6-8 hours on a charge. measurement range— -10 to +10 diopters pupil size: 2mm-8mm
The Quicksee measures roughly 14x10x5 inches and comes with a carrying case. Interview with representative
The device can be fixed in the field by a trained service technician that works for the manufacturing partner. Customer service will be available online. An instruction manual will be provided with the device and the Quicksee will provide feedback on the screen regarding potential misuse if results seem off the mark. The time to train a technician to fix the Quicksee is about 30 minutes to two hours. Interview with representative
Spare parts and batteries will be provided by the manufacturing partner. Interview with representative
- 3-5 years Interview with representative
The Quicksee is designed to approximate the patient’s prescription as accurately as subjective and objective eye exams combined. Plenoptika started with a mission of helping the 1.5 to 2.5 billion people around the world with less-than-perfect vision obtain glasses.
The Quicksee is a medical device. Wrong readings may cause harm to the patient, as they will not receive the care that they need. An instruction manual will be provided with the device and the Quicksee will provide feedback on the screen regarding potential misuse if results seem off the mark. The Quicksee is intended for ages 5+. Interview with representative
N. J. Durr, et al. “Design and Clinical Evaluation of a Handheld Wavefront Autorefractor:,” Optom. Vis. Sci., vol. 92, no. 12, pp. 1140–1147, Dec. 2015.
E. Lage, et al. “Evaluation of a low-cost wavefront aberrometer for measuring refractive errors,” in ARVO Meeting Abstracts, 2014, vol. 55, p. 2718.
E. Lage, et al. “Visual acuity evaluation with refractions prescribed by a novel low-cost wavefront analyzer,” in ARVO Meeting abstracts, 2015, vol. 56, p. 3570.
PlenOptika is working toward U.S. FDA registration for the QuickSee.
refractive errors tests measures sphere cylinder and axis for each eye 10 seconds per eye uses waveform abberometry battery powered device (6-8 hours on a charge) measurement range— -10 to +10 diopters pupil size: 2mm-8mmInterview with representative
In context photographs (0)
You can use the form below to contribute a photo of this product.
In context files & documents (0)
Feedback from users, customers & distributors
Please let us know your experience in using this product through comments below.