Laerdal Upright Newborn Bag-Mask
Laerdal Global Health
This is a manual resuscitator for infants.
The Laerdal Upright Newborn Bag-Mask is a self-inflating, manual, reusable resuscitator intended for newborns and infants up to 10 kg body mass who require respiratory support. This product was selected for inclusion in the WHO’s 2014 Compendium of Innovative Health Technologies for Low-Resource Settings
The 75 countries with the highest burden of maternal and newborn mortality in the world.
The product is being sold on a not-for-profit basis at $20 for use in 75 focus countries with the highest burden of maternal and newborn mortality in the world, also referred to as the UN Millennium Development Goal countdown countries. For those that do not qualify for not-for-profit pricing, the product is also available for purchase in select countries from the Laerdal Medical website for $60-100 USD.
Goal 3: to improve and promote good health.Neonatal asphyxia, a condition that results from oxygen deprivation that is one of the top three killers of newborns globally. About one in ten babies need help to start breathing, and about 90% of these babies can be saved through basic interventions such as bag-mask ventilation.
Physicians, nurses, and midwives.
Upright is manufactured at a plant in China. The product is produced as order requests are received.Interview with representative
According to representatives there are several patents that have been approved for the device. Interview with representative Upright with Newborn PEEP is currently pending US and international patent and design registrations as 2016.
The Laerdal Upright Neonatal Bag-Mask can be purchased on a not-for profit basis for the 75 countries with the highest maternal and neonatal mortality rates. Those that do not qualify for not-for-profit purchase can order the product from Laerdal Medical. Partners in the Helping Babies Breathe Global Development Alliance are using Upright with the Helping Babies Breath resuscitation training curriculum. The products included in the training are typically left with providers upon completion of the program. The product has also been purchased by Ministries of Health.Interview with representative.
Over 5,000 units have been distributed as of May 2016. Upright is used in over 40 countries.Interview with representative
List or none.
Continuous, Recharging only, None, Other
Upright operates like a normal bag-valve-mask device except that the bag has a vertical stance over the mask. The mask fits with standard 15 mm inner-diameter conical connectors and attaches with a snap fit when pressed completely into place. Upright has a pressure release valve (“pop-off ”) which limits the airway pressure to 30-45 cmH2O and makes a hissing sound when opened. A higher pressure can be applied by pressing downwards on the valve with the index finger while squeezing the bag. Testing of lip valve function, pressure release valve, inlet-valve opening, and product sealing must be conducted before use. Procedure for each test are available in Directions for Use.
Expiratory resistance: <2.5 cm H20 at 5 LPM; Inspiratory resistance: <0.5cm H20 at 5 LPM; Tidal volume: > 150 mL; Dead space: 4 mL (water volume)
Operating temperature: -18-50°C; Storage temperature -40-60°C; External dimensions: 72mm x 85mm x 217mm; Mass: 190g
Sterilization method: boiling, chemical glutaraldehyde solution or steam autoclaving
A 1-year warranty is provided under the Laerdal Global Warranty . Laerdal Global Health recommends that customers contact them through their online customer feedback form or email if a product is malfunctioning.Interview with representative
If properly cared for, the product does not require any preventative maintenance. The product can be serviced by a general technician if it needs to be repaired. Depending on the nature of the repair, the manufacturer recommends to purchase a new unit if the bag-mask is not functioning properly because the product is low-cost.Interview with a representative
All spare parts listed in the Directions for Use can be purchased from Laerdal Global Health via email. Supplementary oxygen kits can also be purchased on the Laerdal Global Health website for $4 USD. ^(Interview with representative)
The device may last for years or decades, depending on how it is stored and reprocessed.Interview with representative
Upright is suggested to make obtaining mask seal and providing effective ventilation easier than other newborn resuscitation equipment. The upright orientation of the mask is intended to help the user provide an even, downward pressure onto the mask. The mask has a thicker and broader top surface and a more pliable bottom cuff part to make it easier to hold the mask correctly and thus enabling a better mask seal.
Upright has undergone reprocessing and disinfection testing by NAMSA and passed all acceptance criteria. Upright was tested by PATH in Seattle where results support that the product improves ventilation, is easier to use, and decreases mask leak.1 A study conducted with 87 Tanzanian and Norwegian nursing and medical students revealed similar results.2 Clinical studies in Uttar Pradesh, India and Haydom, Tanzania were recently completed and publications will be available soon.Interview with representative. Future studies are set to take place in Nepal, Uganda, and India.
Laerdal Global Health, NAMSA, PATH, Safer Births, Save the Children, Aligarh Muslim University
Laerdal Upright Newborn Bag-Mask is equipped with a pressure release valve (“pop-off”) that limits the airway pressure to 30-45 cmH20. Instructions are provided in the Directions for Use in the event that the patient valve becomes contaminated with vomit.
The Laerdal Newborn Mask is compatible with the NeoNatalie Resuscitator and other resuscitators. An oxygen kit is available as an optional accessory that can be ordered separately. A Positive End-Expiratory Pressure (PEEP) accessory is currently being developed for Upright and is anticipated to be available by the end of 2016.Interview with representative
Coffey, P.S., Saxon, E.A., Narayanan, I., DiBlasi, R.M. Performance and Acceptability of Two Self-Inflating Bag-Mask Neonatal Resuscitator Designs. Respiratory Care. Published: July 7, 2015. doi: 10.4187/respcare.03867Thallinger, M., Ersdal, H.D., Ombay, C., Eilevstjønn, J., Størdal, K. Randomised comparison of two neonatal resuscitation bags in manikin ventilation. Archives of Disease in Childhood- Fetal and Neonatal Edition. Published: October 5, 2015. doi:10.1136/archdischild-2015-308754
Upright has received CE mark 0434. Meets ISO 10651-4:2002 (Lung ventilators- Particular requirements for operator-powered resuscitators). ANSI/AAMI/ISO 17665, AAMI TIR12,AAMI TIR 30 and ANSI/AAMI ST79 (all regarding reprocessing and disinfection).
Details of evaluations done both in the lab and in clinical settings can be found in the Upright Newborn Bag-Mask Report.
In context photographs (0)
You can use the form below to contribute a photo of this product.
In context files & documents (0)
Feedback from users, customers & distributors
Please let us know your experience in using this product through comments below.