ShangRing Circumcision Device
Jian Zhong Shang
The ShangRing Circumcision Device is a disposable device for male circumcision that eliminates the need for suturing. Results from Chinese studies have demonstrated that the Shang Ring is both safe and easy to use.
Manufactured by Wuhu Santa Medical Equipment Technology Co. Ltd, distributed to health clinics and hospitals. Available for purchase online via Alaibaba
$10-$15 USD via Alibaba.
Surgical circumcision methods, AlisKlamp, SmartKlamp, TaraKlamp, IsmailKlamp, Zhenxi Ring, Prepex
Goal 3: to improve and promote good health.
Male circumcision (MC) reduces the risk of HIV and other sexually transmitted infections (STIs), including human papillomavirus (HPV) and herpes simplex virus type 2 (HSV-2).
Intended for use by trained health care providers for males in geographic areas with high prevalence of heterosexually transmitted HIV and low prevalence of male circumcision in Sub-Saharan Africa.
The device is available at health clinics and hospitals to surgically trained medics. It can also be purchased via online retailer – Alibaba
The ShangRing is a disposable circumcision tool that helps health care providers with limited training to perform circumcision. The device consists of two parts (rings). The outer ring consists of two halves that are hinged together at one end with a ratchet closure. Hemostasis is realized by the pressure applied by the interlocking rings which minimizes bleeding and eliminates the need for sutures. The device must remain on the penis for 5-7 days.
The device is made from plastic, with a weight of 2.5 – 8.5 grams. ShangRign is commercially available in China in 32 sizes, ranging from 9-42 mm in diameter, for use with neonates to adults. The device is pre-sterilized.
The ShangRing must be used by properly surgically trained physicians. Training DVDs, manuals, and curriculum are being developed.
The ShangRing is a single use, disposable device.
The device is sterilised by gamma radiation (Cobalt-60 exposure) and has a shelf life of 3 years. It is strictly non-reusable.
The manufacturer claims that circumcision with the ShangRing is safer and more acceptable to patients than conventional surgical circumcision methods. Additional performance targets include: • No sutures necessary • Rapid procedural time (2 to 5 min) • Less bleeding • Frenulum is well preserved • Low complication rate • Excellent cosmesis • Easy to teach and learn • High satisfaction among medical providers and patients
Results from Chinese studies have demonstrated that the Shang Ring is both safe and easy to use. The first report of the Shang Ring in the literature was from a study conducted by Peng et al. who reported low rates of complication among 1,200 Shang Ring circumcisions. Similarly low complication rates were seen in a series of 328 men circumcised with the Shang Ring in another Chinese study by Cheng et al.. A third study by Li et al. conducted in China comparing Shang Ring circumcision with the dorsal slit technique found significantly fewer complications, lower pain levels during and after the procedure, and higher participant satisfaction among participants circumcised with the Shang Ring. Average Shang Ring circumcision times (excluding administration of local anesthesia), were reported in the Chinese studies to be between 3-5 minutes with patient satisfaction rates over 98. Since 2008, a series of studies using the Shang Ring for adult MC have been carried out in Africa according to guidelines established by WHO for clinical evaluation of new devices for adult MC. Results from the studies conducted to date in Africa demonstrate that the Shang Ring has an excellent safety profile, supporting results of Shang Ring use in China and indicating that the Shang Ring could facilitate rapid roll-out of MC for HIV prevention in Africa. The African experience shows that the Shang Ring is easily used by trained non-physicians, which would facilitate task shifting, and is strongly preferred by providers and well-liked by men who have participated in the studies. The vast majority of participants in studies in Africa as well as China have reported few problems and little disruption to daily life while wearing the device. The significantly shorter procedure time for Shang Ring circumcision would allow for many more procedures to be done in a given time frame compared to conventional MC techniques.
Studies using the Shang Ring for adult MC have been carried out in Africa at a variety of hospitals.
It is strictly non-reusable. Should be only used by surgically trained health care providers.
Clinical trials using the Shang Ring device for male circumcision in Africa: a review Mark A. Barone et.al. EngenderHealth, New York, NY, USA; Center for Male Reproductive Medicine and Microsurgery, Department of Urology, Weill Cornell Medical College, New York, NY, USA; EngenderHealth, Nairobi, Kenya Vol 3, No 1 (March 2014). Submitted Dec 02, 2013. Accepted for publication Jan 24, 2014. doi: 10.3978/j.issn.2223-4683.2014.01.09
Clinical application of a new device for minimally invasive circumcision: Yi-Feng Peng et.al. Asian J Androl 2008; 10 (3): 447–454 DOI: 10.1111/j.1745-7262.2008.00411.x
The ShangRingTM has obtained FDA (510K) clearance in the US, and CE Mark from the European Union and ISO 13485 certification.
Randomized controlled trials tracking user feedback, healing time, blood loss, infection rate and complication rate.
Video presentations of the ShangRing™ surgical technique have been presented in various conferences and received awards for excellence.
In context photographs (0)
You can use the form below to contribute a photo of this product.
In context files & documents (0)
Feedback from users, customers & distributors
Please let us know your experience in using this product through comments below.